O SlideShare utiliza cookies para otimizar a funcionalidade e o desempenho do site, assim como para apresentar publicidade mais relevante aos nossos usuários. Se você continuar a navegar o site, você aceita o uso de cookies. Leia nosso Contrato do Usuário e nossa Política de Privacidade.
O SlideShare utiliza cookies para otimizar a funcionalidade e o desempenho do site, assim como para apresentar publicidade mais relevante aos nossos usuários. Se você continuar a utilizar o site, você aceita o uso de cookies. Leia nossa Política de Privacidade e nosso Contrato do Usuário para obter mais detalhes.
5 single implant case osteotomy prgression in open guide sleeves
# 23 SINGLE IMPLANT CASE Osteotomy Progression in Open Guide Sleeves Featuring 3 x1 mm Magnetic Offset Guide Post Open Guide Sleeve Utilizing Cosine Rule Generation ll Bending Tool GALAXIS Software Evaluaton Invivo5 Software Evaluation Drill series in SURGICAL Guide TUIT
cross sectional view indications Close up NOTE: distance between yellow & green lines = angular correction of 20° GALAXIS necessary to correctly place implant virtual implant should be repositioned 1 mm toward the lingual & positioned in the center of the bone axis should be bent toward the lingual # 23
# 23 Tangential View Indications: Implant could be repositioned 6° Angle bent toward the mesial (left) Implant centered between the roots of the adjacent teeth 6° angular correction is necessary to correctly place implant GALAXIS
ALIGNMENT of guide sleeve with long axis of implant FIRST BEND 20° correction to the lingual # 23 axial cross sectional (bucco-lingual) Invivo5 Anatomage tangential (mesio-distal) volumetric
# 23 Evaluation Plane : yellow line which should be (is) a bucco-lingual orientationInvivo5Anatomage axial view Explanation re: Evaluation Plane see slide show ►“Use of Pythagorean Theorem #9 Single Implant Case” slide # 27
Invivo5Anatomage ALIGNMENT (bucco-lingual) Indication: 20° Angular Correction to the Lingual is needed For FIRST BEND An angle is measured by intersection of the line passing thru the long axis of the implant and the long axis of the guide sleeve. In this case, 20° cross sectional view
3 mm magnetic guide post with 1.0 mm offset in bending block to be bent 20°
Initially, the stylus over the unbent offset guide post should point to zero.
Guide Right™ Products shown in this case 1 - 3/32” drill 1 - 2 mm Guide Post 1 - 2 mm Guide Sleeve 1 - 3 mm X 1 mm offset Magnetic Offset Guide Post 1 - 3 mm Open Guide Sleeve Generation ll ▪ Guide Post Bending Tool Triad® Gel 1.800.314.0065 • www.deplaque.com
Guide Right™ GENERATION II ▪ GUIDE POST BENDING TOOL SINGLE BEND reviewStep 1 Place bending tool plate on a secure flat surface with the degree increments at the top & the stainless steel bar with the v-cut at the bottom.Step 2 Locate 3/32” hole in the center of the v-cut and place the bottom half of the guide post into the hole. Tighten the set screw.Step 3 Locate the hole in the bottom of the stylus that you will use that will fit over the top half of the guide post (3.0 mm, 4.0 mm or 5.0 mm).Step 4 Fit the stylus over the guide post securely with the point directed at zero degrees and the bottom of the stylus in contact with the V block.Step 5 Using the stylus as a lever, bend the guide post to the degree of angle of correction. You may need to ease the point of the stylus beyond the point of the desired degree.Step 6 Loosen screw and remove guide post and the stylus to find the guide post bent to the desired angle.
Guide Right™ GENERATION II ▪ GUIDE POST BENDING TOOL COMPOUND BEND overviewStep 1 Position a straight or offset guide post in the bending plate, tightening the set screw against one of the flat surfaces on the lower half of the guide post.Step 2 The 1st bend can be made to the right or left direction.Step 3 The set screw is loosened and the guide post is rotated 90 ° next flat surface.Step 4 The 2nd bend in the second plane is made after rotating the guide post up away from the surface of the bending plate to register the stylus point back at 0 degrees.Step 5 Slide the stylus support bar down under the stylus until it supports the stylus. Tighten the side screws before making the second bend.Step 6 The second bend can be made in either direction according to the x-ray.Step 7 Remove the stylus and place the guide post back in the cast with the appropriate side indicated by a mark facing the buccal or lingual surface. Be sure the post is in the correct position. If the post needs to be corrected by a linear movement an offset guide post can be used. Off sets available in the 3 mm guide post: 0.5,1,1.5, 2.0 ,or 3.0 mm.
A System of Componentsfor the fabrication and correction of diagnostic & surgical guide in one or two dimensions In-office or lab fabrication Evaluate with 2D & 3D imaging Allows linear and angular correction Enables precision implant placement Cost effective 1.800.314.0065 • www.deplaque.com
Guide Right™ by DéPlaqueStart With Precision. Place With Confidence. 1.800.314.0065 • www.deplaque.com